Adding to the legacy established in previous versions, MassWorks 5.0 is still the easy-to-use post acquisition software package that utilizes Cerno’s patented TrueCal™ technology to achieve high mass accuracy and high Spectral Accuracy and enable elemental composition determination on conventional mass spectrometers of unit mass resolution using the patented CLIPS formula search. MassWorks sCLIPS and BestScan sCLIPS now also provide high spectral accuracy through exact line-shape calibration without the need for standards. MassWorks integrates the powerful TrueCal calibration technology to obtain up to 100X improvement in mass accuracy on unit resolution systems and up to 99.9% spectral accuracy on both high and unit resolution systems in a fast and versatile MS application software package. By mass accuracy with Spectral Accuracy, Cerno methodologies can provide significant improvements to all types of MS data, both high and low resolution.
- High mass accuracy on quadrupole GC or LC/MS (up to 0.005Da)
- Improved Spectral Accuracy on all systems (up to 99.9%)
- Elemental composition determination on quadrupoles
- Eliminating up to 95-99% of incorrect elemental compositions on high resolution MS
- Accurate isotope fine structure analysis on ultra-high resolution MS
- Direct and accurate analysis of unresolved mixtures, e.g.,isotope labeling and biologics modifications
New features for Version 5.0
- NIST Library Search and ChemSpider Integration
The power of GC/MS search coupled with accurate mass (>±10mDa) search and better than 99% spectral accuracy turns your single quad GC/MS into a powerful problem solving analytical tool. Perform accurate mass searches in the NIST library by simply clicking on the spectrum in MassWorks. Verify the library hit with increased confidence using the power of spectral accuracy and the built-in CLIPS formula search. Search formulas from the CLIPS hit list in either NIST or ChemSpider libraries to confirm the compounds and explore their possible structures and other molecular properties.
- BestScan™ sCLIPS
High resolution traps, such as the Orbitrap, can suffer from poor spectral accuracy due to space charge effects even with automatic gain control. MassWorks identifies the best spectrum across the eluting chromatographic peak using Spectral Accuracy to minimize this problem by pinpointing the best MS scan, typically not at or near the chromatographic peak apex, to use for accurate elemental composition determination of unknown compounds.
For a list of all new features click here.
- "The ability to provide the elemental identification of unknown compounds on a routine basis by any scientist is now within one's reach." - Click here to read full presentationEli LilyJ.Mick,T.Gillespie ASMS2007
- "The use of exact isotope modeling [with MassWorks] is key to unique formula ID [with FT-ICR] and profile mode peakshape calibration is needed to achieve the required Spectral Accuracy." - Click here to read full presentationBoehringer IngelheimS.Pennino,F.Qiu,Y.Wang ASMS2008
- "The accurate mass profile extract ion chromatogram (AMPXIC) obtained through accurate mass measurement and isotope profile-mode filtering [performed with MassWorks] eliminated the false positive peaks observed using conventional XIC in the metabolite profile." - Click here to read full presentationWyeth ResearchM.Zhang,M.Gu,N.Kagen ASMS2008
- "Starting a little bit "speechless" from our side your Presentation helped really changing our mindset, from "unbelieving" into "believer". After this Test with a sample taken from real life with real interferences my colleagues and I fully agree that MassWorks is indeed a powerful tool for advanced GC/MS analysis. We found that MassWorks is really a "must have" product for us..."Researcher from a large German chemical company
- "Congratulations!! Once more I are am deeply impressed by sClips. In both cases you are right with the top no. 1 [formula]!! Additionally, I find it fantastic that you do not need a calibration standard for higher resolving mass specs, by “just” looking at the isotope signals."- Communication from blind sample evaluation on a Leco LC-TOFUniversität Hohenheim, GermanyProf. Dr. Wolfgang Schwack
- "The program is very easy to learn, it literally took minutes to get the first exciting results. On our 10 year-old triple quad, we manage to achieve mass accuracies of 10 ppm or better, routinely! It has become an integral part of our characterization pipeline for newly synthesized compounds."University of ViennaDr. Alexander Leitner